Soft pliable rubber like material, best suited for fabrication of bleaching splints. Material does not bond to acrylic. Hardness Shore A: 92.
My splint got a milky appearance after thermoforming. What is the reason for this?
Please make sure to apply BIOPLAST® insulating agent prior to thermoforming. The insulation should still be wet while the thermoforming process is being carried out. In addition, you can also briefly water the model before applying the liquid insulation. However, if you intend to make a bleaching splint using BLUE-BLOKKER® as space holder, the milky appearance might be useful: The clear material areas above the space maintainer may later serve as an orientation for the patient when applying the bleaching liquid. In this case, it is sufficient to water the model prior to thermoforming.
What should plaster models for pressure moulding applications look like?
Plaster models should be finished so as to ensure a good fit of the splint. Therefore take care to block out cavities on the model (due to incorrect casts) with plaster, silicone or resin. Excess plaster, that may be close to the cervical rim or in interdental spaces, should be removed with an instrument. Use a trimmer to adjust the height and rims of the model and smooth sharp edges with a plaster knife or bur. Block out undercuts prior to pressure moulding, using preferably light-curing blocking-out resins or silicones, such as BLUE-BLOKKER® or SIL-KITT. Do not use waxes, as they might melt when getting in contact with the heated thermoforming materials. If you want to thermoform hard materials, teeth might break when removing the material. Therefore it is recommended to use duplicate models. Press an ISOFOLAN® foil for insulation over the model, cut the foil at the model base and perforate it interdentally. Embed the model into the pellets to prevent the material from overstretching. The model should be placed with the front teeth showing in the direction of the pivot axis or the locking shaft to ensure optimum adaptation of the material. If you want to thermform soft materials, it is recommended to use preferably flat models (without guiding grooves). Position your isolated model on the model platform or embed it into the pellets using the pellet cover with the front teeth showing in the direction of the pivot axis.
Is it necessary to isolate the model prior to thermoforming using a foil?
That depends on the type of material you want to press over the model. Recommendations for model insulation can be found in the instructions for use of the material and in the indication chart on applications in the pressure moulding technique. Both documents can be downloaded on this page under "Further Information".
When should the model be placed on the model platform and when should it be embedded into the pellets?
Basically, you should avoid overstreching the material and thus reducing the material thickness. If you are using a dental arcade model, the use of the model platform is recommended. If you are using a full model, it should be embedded into the pellets. For soft materials we recommend using the reusable pellet cover, preventing the pellets from sticking to the heated material and reducing loss of pellets.
Which material side has to face to the top when fixing the material in the thermoforming unit?
Make sure to fix the material with the imprint of the clamped material being readable from above. This applies to all SCHEU-DENTAL pressure moulding devices. If you are using devices from other manufacturers, please verify the heating system, as with some systems the material is not pivoted after being heated. When using material without imprint/codes on it, the side you are fixing the material does not matter.
What happened to my splint not being properly adapted to the model after thermoforming?
Maybe too much pressure has been applied when removing or cutting and finishing the splint.
I’m using the same material, but it has a different code printed on it. Why is that?
With each new material production, the processing parameters of the material, such as heating and cooling time, are reviewed by our technicians and adjusted if necessary. So there might be minimal deviations from batch to batch, offering you, the user, even more reliability when processing the material.
Can I use the material again if I have failed to initiate the thermoforming operation and the infrared heater has already been automatically switched off?
No, because the prolonged exposure to heat, humidity and oxygen may affect the material properties. Please dispose of the material, start a new heating process using new material and take care to thermoform the material immediately after the end of the heating phase. Both the BIOSTAR® and the MINISTAR S® pressure moulding units indicate the end of the heating phase by acoustic signals.
SCHEU-DENTAL offers you a wide range of orthodontic products.
High-quality dental technical supplies for your everyday practice.
Use innovative procedures to treat your patients.
Put your trust in the market leader for dental sleep medicine.
Any questions? We can help.
|Mo-Thu||7:45 am - 4 pm|
|Fr||7:45 am - 2 pm|
We regularly research and develop new ideas that make your everyday work in the laboratory and practice easier.